Guide autoclave with vertical loading; gravity displacement type with possibility for Vacuum.
Capacity: Up to 12 kg/h
Chamber size: >150 Dimensions
Material: Stainless steel
Temperature: 134 °C
Minimum autoclave room size shall be 150 liters and shall have a usable quantity (loading basket) of 100 liters.
Shall work based on gravity air displacement principle.
The vacuum system shall be of a thermal shock kind.
The pressure vessels ought to be pressure tight, the rising of the pressure (at a start pressure of below 80 mbar)
shall be lower than one mbar/min.
With in-build electrical heating elements for steam generation (heating elements > 4000 Kw).
Chamber and door shall be made from high quality stainless steel (e.g. AISI 3161).
Chamber in vertical position, it shall enable a simple loading/unloading of this loading basket. The baskets shall be made for the
usage of non-heat resistant waste bags, a system for the collection and controlled release of free liquids and
condensate in the waste shall be integrated.
The temperature measured at the reference measurement stage of the treatment room shall not be lower than 105°C
(TDes 0+4). The treatment time shall be ≥30 minutes at 105°C, ≥20 minutes using a temperature used of ≥121°C (TDes
0+4) and ≥10 minutes using a treatment temperature of ≥ 134.
The steering of the treatment process should be semi-automatically in addition to manual.
The Treatment Equipment shall be equipped with the following instruments:
o treatment room temperature indicating instrument
otreatment room pressure indicating instrument
For the treatment of the outgoing air, a HEPA filter in combination with a condensation system should be used.
The filter shall be constructed in a manner that a disinfection of the filter shall be possible in an integrated condition.
The filters must be heat resistant, steam resistant, hydrophobic and pressure resistant for the pressure difference.
Cooling down phase after treatment ought to be existent; the temperature of the waste should be not higher than 80°C
after completing the cooling down process.
The pressure vessel should be equipped with a heating insulating material, the maximum outer surface temperature should be < 50°
after finishing the cooling down process.
The pressure vessel should be equipped with a heat insulation, the maximum outer surface temperature should be < 50°